
Pt 100 foil elementPt100 Foil element Art.nr: 263601 [Larger] Technical data
Temperature range
200...+200°C. Response time T0,5 = 1 sec on surface. Measurement uncertainty According to DIN IEC751 class A: 0,15 + 0,002t* °C *t = the absolute value of the temperature in degree Celsius. General Pt 100, 11x30x0,13mm
The resistance of a Pt100 sensor at 200  0°C is calculated according to this formula: Rt=R0(1+3,9083x103t  5,775x107t2  4,183x1012(t100)t3) The resistance of a PT100 sensor at 0  850°C is calculated according to this formula: Rt=R0(1+3,9083x103t  5,775x107t2) R0, resistance at 0°C. (100 Ohm) Rt, resistance (Ohm) at the temperature t. (°C) 


